An automated ground facility, for vicarious radiometric calibration of airborne and spaceborne sensors of visible and infrared light has been established. In the term "vicarious calibration," "vicarious" is used in the sense of "in place of another," signifying "in place of laboratory calibration." Vicarious calibration involves the use of ground truth in the form of measurements by ground-viewing radiometers, a Sun-viewing photometer, and meteorological instruments positioned in a ground target area. The instrumentation at the facility includes a light-emitting-diode spectrometer (LSpec), which consists of eight tripod-mounted, ground-viewing radiometer units containing LEDs biased to operate as photodetectors (instead of light emitters) at their respective wavelengths. The LSpec provides an essentially continuous stream of measurements at eight discrete wavelengths. These are merged with spectral surface-reflectance measurements made on occasional site visits to obtain temporally continuous coverage with high spectral resolution.